Development of a biosensor protein bullet as a fluorescent method for fast detection of Escherichia coli in drinking water
Development of a biosensor protein bullet as a fluorescent method for fast detection of Escherichia coli in drinking water
Ignacio GutieÂ rrez-del-RÂõo 0 1
Laura MarÂõn 0 1
Javier FernaÂ ndez 0 1
MarÂõa AÂ lvarez San MillaÂ n 0 1
Francisco Javier Ferrero 1
Marta Valledor 1
Juan Carlos Campo 1
Natalia CobiaÂ n 1
Ignacio MeÂ ndez 1
Felipe LomboÂ 0 1
0 Research Group BIONUC, Departamento de BiologÂõa Funcional, AÂrea de MicrobiologÂõa, University of Oviedo , Oviedo, Principality of Asturias , Spain , 2 Department of Electric , Electronic, Computer and Systems Engineering , University of Oviedo, Campus of Gij oÂn, GijoÂn, Principality of Asturias , Spain, 3 HIPSITEC S.A., Oviedo, Principality of Asturias , Spain
1 Editor: Chien-Sheng Chen, National Central University , TAIWAN
Data Availability Statement: All relevant data are
within the paper and its Supporting Information
Funding: This work received support from FICYT
IE09-106 and IDEPA IDE/2012/000452. NC and IM
belong to HIPSITEC SA and were taking part in this
study in the following activities: NC in the
conceptualization, investigation (construction of
some of the plasmids for expression of the magic
bullet biosensor proteins in S. albus), methodology
Drinking water can be exposed to different biological contaminants from the source, through
the pipelines, until reaching the final consumer or industry. Some of these are pathogenic
bacteria and viruses which may cause important gastrointestinal or systemic diseases. The
microbiological quality of drinking water relies mainly in monitoring three indicator bacteria
of faecal origin, Escherichia coli, Enterococcus faecalis and Clostridium perfringens, which
serve as early sentinels of potential health hazards for the population. Here we describe the
analysis of three chimeric fluorescent protein bullets as biosensor candidates for fast
detection of E. coli in drinking water. Two of the chimeric proteins (based on GFP-hadrurin and
GFP-pb5 chimera proteins) failed with respect to specificity and/or sensitivity, but the
GFPcolS4 chimera protein was able to carry out specific detection of E. coli in drinking water
samples in a procedure encompassing about 8 min for final result and this biosensor protein
was able to detect in a linear way between 20 and 103 CFU of this bacterium. Below 20
CFU, the system cannot differentiate presence or absence of the target bacterium. The
fluorescence in this biosensor system is provided by the GFP subunit of the chimeric protein,
which, in the case of the better performing sensor bullet, GFP-colS4 chimera, is covalently
bound to a flexible peptide bridge and to a bacteriocin binding specifically to E. coli cells.
Once bound to the target bacteria, the excitation step with 395 nm LED light causes
emission of fluorescence from the GFP domain, which is amplified in a photomultiplier tube, and
finally this signal is converted into an output voltage which can be associated with a CFU
value and these data distributed along mobile phone networks, for example. This method,
and the portable fluorimeter which has been developed for it, may contribute to reduce the
analysis time for detecting E. coli presence in drinking water.
(designing in part of the analyzer device for optical
excitation and signal detection of the bound
proteins to bacteria), supervision (of the
engineering part of the project) and funding
acquisition (proposals coordination and
submission). IM in the methodology (tests with the
device for checking signal detection parameters)
and software (for controlling the
Competing interests: HIPSITEC SA received public
funding from the public agencies FICYT (FundacioÂn
para el Fomento en Asturias de la InvestigacioÂn
CientÂõfica Aplicada y la TecnologÂõa) and IDEPA
(Instituto de Desarrollo EconoÂmico del Principado
de Asturias) through the Projects IE09-106 and
IDE2012/000452 for carrying out these research
activities, at its R&D Department (NC, IM), together
with the University of Oviedo at the Department of
Functional Biology (IGR, LM, JF, MASM, FL) and
Department of Electric, Electronic, Computer and
Systems Engineering (FJF, MV, JCC). This
commercial affiliation does not alter our adherence
to all PLOS One policies on sharing data and
materials. Authors declare absence of competing
Water is an essential resource for all life forms and its quality is under constant challenge at
water treatment plants and along transportation pipes towards consumerÂs tap. Throughout
this journey, and especially in its source, water is exposed to an increasing number of
infectious, chemical and radioactive pollutants that can be ingested by humans, causing serious
health problems .
Therefore, drinking water may, in certain cases, be the cause of disease in humans. The
reason is that water can act as a vector of a large number of environmental toxic compounds,
such as toxins of microbial (cyanobacterial hepatic toxins) or mineral origin (mercury,
radioactive compounds). These toxins may be derived from human activities, as poisons of
industrial origin (dioxins, organotin compounds). Furthermore, drinking water can also be the
gateway to our organism for different pathogens, such as parasites, fungi, bacteria, or viruses;
by means of faecal contamination or improper water treatment [
]. Countries such as Norway
or the USA have described massive outbreaks of intestinal parasites such as Giardia,
Cryptosporidium and Entamoeba, due to problems along the networks of water supply [
Canada, Toxoplasma infections cause neurological damage in newborns whose annual treatment
costs around 200 million € [
]. Bacteria with high mortality rates like Leptospira o Legionella
are transmitted by water distribution networks with poor maintenance, affecting hundreds of
]. In tropical countries, cholera cases, transmitted by consumption of contaminated
water, may increase to hundreds of thousands during outbreaks . In the case of viruses, the
most common one transmitted by contaminated water is hepatitis A virus, with about 1.5
million people affected yearly [
]. Therefore, access to clean drinking water is essential, leading to
significant health benefits [
As a consequence, contamination of drinking water with bacteria and other pathogens
remains a major cause of health problems in industrialized and developing countries . The
WHO/UNICEF Joint Monitoring Program for Water Supply and Sanitation estimates that
663 million people lack access to safe drinking water sources , this being a leading cause of
death worldwide (only after respiratory infections and AIDS) with 3.4 million deaths annually
[13±15]. It is estimated that 4% of annual deaths and 5.7% of annual morbidity are caused by
waterborne diseases, and some areas are particularly vulnerable to waterborne diseases, due to
their underdeveloped infrastructure and lack of enough water treatment plants. However, the
distribution of drinking water and wastewater in high-income areas also require monitoring
of microorganisms and contaminants . In fact, in countries like USA, it has been estimated
that 7.1 million people contract waterborne infections each year, of whom 12,000 die annually
To ensure its safety, drinking water must meet diverse requirements, such as absence of
pathogens and certain chemicals, as well as being tasteless, colourless and odourless .
However, it is impossible to sample every hour at each point along distribution pipes, and it is not
feasible to detect all possible pathogens or toxins in these assays. So most countries have
established a consensus protocol by which, once or few times per day, drinking water is tested in
search for the presence of the main chemical toxins (heavy metals, pesticides, etc.) and a few
microbial indicators. The absence of these bacterial indicators is considered to be a proof of
good microbial quality, as they are sentinel species of faecal origin. The most commonly used
bacterial water indicators are Clostridium perfringens, Enterococcus faecalis and Escherichia coli,
which must be absent (for example, in 100 mL water samples) in order to assure that any
drinking, recreational or farming water is safe for humans [18±21]. E. faecalis is mainly used as
an additional microbiological indicator when an E. coli single analysis is not suitable because
of the suspicion of its possible multiplication in tropical waters . On the other hand, C.
2 / 20
perfringens is barely used as a unique microbiological indicator because of its extremely long
half-life . For all of these reasons, Escherichia coli was chosen in 1890 as the main reference
indicator to assess the microbiological quality of drinking water, as it meets all the conditions
for a good sentinel bacterial: it is present in all mammals faeces in large amounts, it hardly
multiplies outside the host, and it is detectable by sensitive methods that require simple
services at a bacteriological laboratory [23,24].
A common method considered to be the gold-standard for detection of these indicator
bacteria is its growth on selective agar media for enterobacteria, which contains inhibitors for
preventing growth of other unwanted bacteria. However, these traditional culture methods used
for quantification of indicator bacteria require days to obtain results. Therefore, there is an
urgent need for the development of a cheap, fast and reliable method to detect E. coli in
drinking water [11,25]. Water safety would increase with the development of faster and portable
methods for E. coli detection, allowing multiple sampling sites and many inexpensive tests per
day. These real time biosensors would therefore diminish the lapse of alarm time between any
eventual contamination along distribution pipes and the health authority's response. Current
developed methods for fast detection include DNA hybridization, PCR (and qRT-PCR for
quantification), immunoassays, immunomagnetic separation, lateral flow tests, incubation
micro-chambers as VITEK, labelled nanoparticles, NIRS (near infrared spectroscopy),
DEP-FFF (dielectrophoretic field-flow fractionation) and β-D-glucuronidase assays [26±34].
But these current technologies are not inexpensive neither in equipment (as NIRS) nor in
reagents (like labelled antibodies), and usually require an intense manipulation of water
samples and laboratory equipment (as DNA-hybridization), which impairs its portability as well as
the feasibility of testing every few minutes. Another low-cost and easy to handle technologies
have been proposed for the detection of pathogenic bacteria, including colorimetric,
electroluminescence, immunomagnetic detection and electrochemical methods. Moreover, recent
advances in nanotechnology have enabled the development of new technologies based on
nanoparticles, nanorods, nanosheets and 3D-nanostructures for rapid and sensitive detection
of pathogens [
], but most of these methods are still very expensive for commercial use and
require sophisticated instrumentation .
However, in recent years an intensive research has been conducted to simplify these
technologies, generating portable biosensors with immediate in situ results. These detection
biosensor assays must also be both sensitive (able to detect low concentrations of the target agents
without the interference from other water particles) and specific (for example, for E. coli and
not for other enterobacteria) [
]. In this work, we describe the development of a portable
biosensor system (578 g) for fast (8 min) and sensitive (10 CFU/L) detection of E. coli in
drinking water samples, with a linear range for over 20 to 103 CFU. Three different approaches have
been made, in order to develop these chimera proteins as specific and sensitive biosensors for
E. coli in drinking water, however, only one of them, containing a GFP domain linked to a
colicin S4 bacteriocin domain, has performed the required binding capabilities to the bacterial
cells. The low cost of this equipment and its automation capabilities make this biosensor a real
possibility to monitor multiple places along water distribution, eventually sending this
information in real time to central control offices, for example by using a mobile phone network.
The main objective of the method described in this study is to control the quality of water in a
cheap way and in multiple locations throughout the distribution system as well as to rapidly
process samples and to share real-time test results.
3 / 20
Material and methods
Bacterial strains, plasmids and culture conditions
E. coli TOP10 F' (Invitrogen, a strain with wt genes coding for for Fhu and OmpW), and
cloning vectors pUC57 (GenScript) and pIAGO [
] were used for routine sub-cloning while
Streptomyces albus was used as host for heterologous protein expression. E. coli clones were
incubated at 37ÊC and 250 rpm in TSB (Tryptic Soy Broth, Merck) or LB (Luria-Bertani
Broth, Merck) for growing in liquid medium, and TSB containing agar was used for growing
on solid medium. Appropriate concentrations of antibiotics were added for plasmid selection:
100 μg/mL ampicillin for both vectors. S. albus was sporulated on Bennet medium [
] at 30ÊC
and grown in liquid YEME medium for protoplasts preparation [
]. Transforming colonies
were cultured at 30ÊC on R5 solid medium, sporulated on Bennet medium and subcultured in
R5A liquid medium [
] for protein expression. Spores were kept in glycerol 25% at -20ÊC.
Media were supplemented with thiostrepton at 5 or 50 μg/mL (liquid or solid cultures
respectively). Enterobacter cloacae (strain CECT 194T) and Salmonella enterica var. arizonae
(strain CECT 4395) were used as negative controls, and have been grown in TSB medium as
Plasmid DNA extraction, restriction endonuclease digestion, DNA ligation, transformation
and other DNA procedures were performed according to standard techniques for E. coli [
Preparation of S. albus protoplasts, transformation and selection of positive transforming
colonies were carried out as described [
]. Restriction enzymes, buffers and T4 DNA ligase were
purchased from EURx. The three chimera proteins and the ermE -rbs promoter were obtained
by gen synthesis (GenScript).
pb5 phage protein (accession number AAU05292.1) was chosen as a sensor bullet for the
detection of E. coli. The synthetic gene coding for Green Fluorescent Protein (GFP) (accession
number AAA27722) was fused to the synthetic gene coding for pb5 protein (640 amino acids).
At the 5' end of GFP gene, a polyhistidine-tag coding sequence was fused in order to facilitate
further protein purification. A synthetic DNA coding for a flexible peptide linker with the
structure of a random coil was designed between GFP and pb5 domains, (ELFLRSLQDKYSN
QRGHYPLLNNPEATNAGKYEYWPSH). This encoded flexible linker contains neither α-helixes
nor β-sheets (tested with ExPASy tools: CFSSP, HMMTOP and TMHMM), and therefore
facilitates anchoring of pb5 bullet domain to E. coli outer cell structures, without hindering
GFP function. The gene coding for the biosensor chimera protein GFP-pb5 (accession number
KY404234) was synthesized and codon-optimized for Streptomyces expression, adding extra
restriction enzyme sites for cloning purposes.
Hadrurin peptide (accession number NDB21_HADAZ), from a scorpion venom, was
chosen as another sensor bullet protein for the detection of E. coli. A polyhistidine-tag sequence
was designed, fused to GFP (accession number AAA27722) and this to hadrurin. The same
linker between GFP and hadrurin was used, and a synthetic gene coding for a biosensor
chimera GFP-hadrurin (accession number KY404233) was also synthesized and codon-optimized
for its expression in Streptomyces, adding extra restriction enzyme sites for cloning purposes as
it has been said before.
Colicin S4 bacteriocin protein (accession number CAB46008) was chosen as the third
sensor bullet protein for the detection of E. coli. For this third biosensor chimera protein, it was
also fused to GFP (accession number AAA27722) and to a polyhistidine-tag sequence. The
4 / 20
same linker was used as well. The synthetic gene coding for the biosensor chimera GFP-S4
(accession number LT548289) was synthesized and codon-optimized, adding extra restriction
enzyme sites. All three synthetic genes contained a rbs for use in Streptomyces.
Finally, in order to improve protein expression, the strong ermE promoter (accession
number M11200, PermE ) was used as well. This promoter allows a strong constitutive
expression of the protein in Streptomyces genus.
Construction of three expression plasmids
pUC57 vector, containing GFP-pb5 synthetic gene construction (2,967 nt insert,
pJFF13-GFT5) was digested XbaI-HindIII and the resulting fragment was subcloned into the
bifunctional multicopy vector pIAGO, giving rise to plasmid pJFF -BS2-NP. The ermE
promoter present in pIAGO vector ensures the proper translation of the cloned gene, generating
the final construction named pJFF1 -BS2-NP. This plasmid was sequenced to confirm its
identity and transformed into S. albus protoplasts.
Also, pUC57 vector containing GFP-hadrurin synthetic gene construct (1,173 nt,
pMAS12-GFHD) was digested XbaI/HindIII and the resulting fragment was subcloned into
pIAGO, giving rise to plasmid pMAS40 -BS2-NP. This plasmid was sequenced to confirm its
identity, and transformed into S. albus protoplasts.
Finally, pUC57 vector containing GFP-colicin synthetic gene construct (2,757 nt,
pMAS11-HPS4) was digested XbaI/HindIII and the resulting fragment was subcloned into
pIAGO, giving rise to plasmid pLMF -BS2-NP. This plasmid was sequenced to confirm its
identity and transformed into S. albus protoplasts.
Protein extraction and purification
Spores of S. albus (107 spores/mL) harboring pJFF1 -BS2-NP, pMAS40 -BS2-NP and pLMF
BS2-NP were grown in 25 mL of TSB medium as preinoculum. Thiostrepton was added as
selective antibiotic and the culture was grown overnight at 30ÊC and 250 rpm.
Each one of the three preinocula was used for inoculating two 2500 mL Erlenmeyer flasks,
each one containing 250 mL R5A culture medium supplemented with glycine (0.5% final
concentration) and thiostrepton. These production cultures were grown for 2 days at 30ÊC and
250 rpm. After centrifugation, cell pellets were washed twice with 10.3% sucrose and then
resuspended in PBS buffer with 2 mg/mL lisozyme plus EDTA-free protease inhibitors mix (Thermo
Scientific) and incubated for 60 min at 4ÊC. Protoplasts formed after this lysis step were broken
by sonication (4ÊC, 4 pulses of 30 s, 1 min stop between pulses). After sonication, cell solutions
were centrifuged at 15,557 rcf for 10 min at 4ÊC in order to get rid of cellular debris.
Supernatant was then used for protein purification by FPLC in two steps. First, a 5 mL
HisTrap column (GE Healthcare) was used for purification of His-tag GFP-linker-bullet
biosensor chimera proteins. Elution buffer A was 20 mM imidazole, 0.2 M sodium phosphate
buffer pH 7.1 and 0.5 M NaCl; elution buffer B was 500 mM imidazole, 0.2 M sodium
phosphate buffer pH 7.1 and 0.5 M NaCl. A gradient was applied between buffers A and B with a
2.5 mL/min flow rate during 40 min. Fluorescent fractions from this first column purification
step were combined and concentrated in order to change the buffer to PBS buffer with
EDTAfree protease inhibitors (Thermo Scientific). These protein concentration steps were carried
out with Pall Macrosep Advance centrifugal devices 10K MWCO (for GFP-hadrurin biosensor
chimera) or 30K MWCO (for GFP-pb5 and GFP-colicin S4 biosensor chimeras).
Concentrated protein solutions (1 mL final volume) were then further purified with Superdex
200 10/300 GL column (GE Healthcare), as a size exclusion second purification. Purified
proteins were quantified by using Bradford and stored at 4ÊC in PBS with protease inhibitors.
5 / 20
Electronic measurement system
A portable measurement device has been developed for detection of E. coli cells, once these
cells may have been labelled with the corresponding fluorescent biosensor chimera protein.
The construction of this E. coli Analyzer device has been already described [
]. The rationale
of the measurement is based on the detection of the fluorescence emitted by the GFP protein
domain immobilized in E. coli cells, via the linker-bound bullet protein domain of each
designed biosensor chimera. The system comprises a light source which excites the fluorescent
chimera protein (Fig 1). Then, the light emitted by this biosensor chimera protein (bound to
E. coli) is filtered before reaching the photomultiplier optical detector [
]. The light signal is
finally converted into an electric signal output (Fig 1). The power supply allows the instrument
to be charged from the power grid or by rechargeable batteries for applications on-site,
allowing a convenient portability of the device.
The optical subsystem in the device consists of the light source, the measuring cell (for
getting 2 mL cuvettes) and the light detector. This design was carried out on the basis of the
excitation and emission spectra of GFP protein, where the excitation peak of greatest intensity is at
Fig 1. Functional blocks of the measurement system at the portable E. coli Analyzer device.
6 / 20
395 nm. Therefore, this wavelength is the corresponding to the peak of the light source
emission (LED diode) located in the portable device [
Exposed to this LED excitation source, the GFP domains of the biosensor chimera proteins
emit a fluorescence whose peak is at 509 nm. Therefore, the light detector incorporated in the
E. coli Analyzer portable device is highly sensitive to this wavelength.
Finally, the device microchip microcontroller reads the value of fluorescence intensity
coming from the photomultiplier and it converts this intensity to an output mV (Fig 1). It also
adjusts the value of the gain of the PMT (via touch display), manages the process of calibration,
The calibration process is performed before each sample measurement. This step is critical
in order to obtain good results. For this, a sample of PBS buffer containing the corresponding
E. coli dilution (blank sample) is placed into the measuring cell and the output voltage read
under these condition is stored. If the system were ideal this voltage should be zero volts, but
the photomultiplier generates a certain offset voltage. Then, the real water sample to be
analyzed is placed in the cuvette, and the control system reads the voltage value obtained and
subtracts the offset voltage. Finally, interpolating this value in the calibration curve, the number of
CFUs is obtained and it is shown in the display.
A single colony of E. coli Top10 was inoculated in 5 mL of LB medium and incubated at 37ÊC
until reaching an OD of 0.8 measured at 600 nm with a Biophotometer (Eppendorf). Decimal
serial dilutions from this culture were done in PBS buffer (from 3 to 3x105 CFUs/mL).
2 mL of each purified biosensor chimera protein solution were incubated with 100 μL of
the corresponding E. coli dilution for 5 min (room temperature, 100 rpm vortex). This mixture
(2100 μL) was then filtrated using a 0.2 μm cellulose acetate filter cartridge (Whatman). In
order to eliminate the unbound biosensor chimera protein on the filter membrane, 2 mL of
extra PBS buffer was used as a wash step.
At this point, E. coli cells (with and without bound biosensor chimera proteins) were eluted
with 2 mL of PBS buffer, using the same filter, but in the reverse position; this step allowed to
recover all cells present on the filter. Fluorescence was measured in 2 mL fluorimeter cuvettes
(VWR) using this 2 mL recovered solution, and this measurement was carried out with the
developed E. coli Analyzer device. Blank solution for these measurements was PBS buffer with
the corresponding E. coli dilution and protease inhibitors, these PBS blank samples were
measured before each experimental sample in order to eliminate putative background
interferences. This device registered for each binding assay a mV value which corresponds to the
equivalent fluorescence emitted by the biosensor chimera protein bound to the E. coli cells
present in the experiment, and this mV value is automatically referred to a CFUs measure
which was established by using a gold standard test. This test was carried out by plating out
each dilution on EMB agar plates and incubating them overnight at 37ÊC [
Images were collected with Leica TCS-SP2-AOBS and Leica TCS-SP8X confocal microscopies
(Servicios CientÂõfico-TeÂcnicos, Universidad de Oviedo) using a 63x/1.40 Oil objective. GFP
was excited using a 488 nm argon/krypton ion laser and a white laser, and fluorescence
emission was detected at 502±556 nm. Leica Confocal Software (LCS) version 2.61 Build 1537 and
Leica Application Suite X (LASX) were used for data acquisition (Leica Microsystems
Heidelberg GmbH (Germany)). S. albus transformation clones for pJFF1 -BS2-NP, pMAS40
BS2-NP and pLMF -BS2-NP plasmids were cultivated in R5A liquid medium, and these
7 / 20
mycelia were analyzed by confocal microscopy in order to detect production of the
corresponding fluorescent biosensor chimera protein. E. coli samples used for binding assays of
each chimera fluorescent protein were analyzed under confocal microscope as well, following
the same protocol that has been described in the ªDetection assayº section.
Also, in order to determine if binding of the fluorescent biosensor chimera protein to E. coli
cells surface was stable in each of the three cases, the recovered E. coli solution (after chimera
binding) was filtered again before a second round of confocal microscopy studies.
Three proteins with binding capabilities to E. coli have been chosen and analyzed as biosensor
alternatives for this system. The first option was hadrurin, a small protein (41 amino acids)
from the venom cocktail of the Mexican scorpion Hadrurus aztecus [
]. The second biosensor
protein chosen was the tail protein pb5 from T5 phage, which is the one interacting with the
receptor protein at the outer membrane of E. coli [
]. Finally, the third alternative was colicin
S4, a bacteriocin recognizing E. coli cells, which was actually the only one providing good
performance in the detection tests [
GFP-Hadrurin biosensor chimera protein
Hadrurin is an antimicrobial peptide (AMP) isolated from the venom of Hadrurus aztecus
scorpion. AMPs have a broad spectrum of target bacteria and induce cell death in a short
contact time [
]. This antibacterial peptide is composed of 41 amino acids with a molecular mass
of 4.4 kDa. Its antimicrobial effect is present at low concentrations, in a micromolar range,
inhibiting the growth of a wide range of bacteria, including E. coli. A specific receptor for the
action of hadrurin has not been described, and the most likely mechanism of action would be
by destabilization of the cytoplasmic membrane activity due to formation of transient pores
The hadrurin peptide was selected as a first candidate for testing it as a sensor protein
targeting E. coli cells in our method (Fig 2). Therefore, S. albus-pMAS40 -BS2-NP spores were
cultivated in 250 mL R5A production medium in order to get cell mass to isolate
GFPhadrurin biosensor chimera protein (386 amino acids, Fig 2A). After FPLC purification
steps, a concentration of 152 μg/mL sensor bullet was obtained (measured by Bradford). 2 mL
of GFP-hadrurin protein were used for binding assays to 100 μL of each E. coli dilution (see
Materials and methods section).
In order to visualize binding of the GFP-hadrurin biosensor chimera protein to cells, the
fluorescence was measured with the E. coli Analyzer fluorimeter (see Materials and methods
section). However, these experiments showed low fluorescence signals.
We decided to analyze the same samples under confocal microscope, with and without the
filtering step used to get rid of unbound biosensor chimera protein after mixing the protein
with the E. coli dilutions. Confocal images showed that GFP-hadrurin labeling of cells was very
low once the labeled cells were filtrated (Fig 3). As this GFP-hadrurin low labeling resulted in
low signals, no further experiments were carried out with this chimera.
GFP-pb5 biosensor chimera protein
The infection process of Gram-negative bacteria by a bacteriophage begins with a highly
specific mechanism of interaction in which the receptor binding proteins (RBP) of the phage bind
particular structures exposed on the surface of the bacterium. The RBP pb5 (67.8 kDa) from
phage T5 is located at the tip of the central tail fibers, near the baseplate, and ensures binding
of phage T5 to its receptor, placed on the outer membrane of E. coli cells. This receptor is the
8 / 20
Fig 2. A: Hypothetical spatial structure of the fluorescent biosensor chimera protein GFP-hadrurin. B: Diagram showing the attachement process of the
fluorescent biosensor chimera protein to the membrane of E. coli. PMT: photomultiplier (part of the E. coli Analyzer device in charge of detecting and amplifying
the fluorescence signal produced by the GFP domain of each biosensor chimera protein, after receiving the 395 nm excitation light from the LED source).
outer membrane iron-ferrichrome transporter FhuA, which mediates the transport of iron, as
it is usually present in very low concentrations in the environment [
As another approach in order to develop a biosensor chimera protein for detection of E. coli
in drinking water samples, a second synthetic gene was designed, in which the GFP domain
was linked to the already described pb5 RBP from phage T5 (Fig 4).
This pb5, structurally, in comparison with other RBPs, it shows a high content of β-sheet
and a low helical content [
]. The crystal structure of its receptor on the bacterial outer
membrane, FhuA, comprises a 22 β-strands barrel that is inserted in the outer membrane of the
bacteria and contains a plug which fills the lumen of the barrel. This plug is a globular N-terminal
domain which penetrates the barrel from the periplasm side and closes the pore of the barrel
]. The plug has four-stranded short β-sheet and four short helices that are
connected both to the barrel and to hydrophilic loops facing the external medium . Pb5 binds
to two large external loops located in the FhuA barrel [53±56].
The in vitro interaction between purified pb5 and FhuA generates a highly stable
stoichiometric complex, which possesses the same stability as in the case of the complete phage
[53±55,57]. The fact that a purified soluble pb5 protein is able to bind its receptor forming a
stable complex is not obvious, because the binding proteins of the phage tail undergo several
conformational changes when binding its bacterial receptor, which are transmitted to
neighbouring tail proteins, initiating a series of conformational changes that convert phage tail into
an appropriate structure to be crossed by DNA. Taking into account this mutual
interdependence in relation to conformational changes of proteins in the phage tail, one could think that
9 / 20
Fig 3. Confocal microscopy images of E. coli (A±G, a±g), Salmonella enterica (H, h) and Enterobacter cloacae (I, i) fluorescent labeling tests. Each group of 4
images correspond to a different chimeric biosensor fluorescent protein. Images from ªAº to ªIº correspond to confocal microscopy, showing the fluorescent
labelled cells, and images from ªaº to ªiº are the same confocal images but fused with an optical image generating a single photo (merged snapshot for testing that
fluorescent spots actually correspond to labelled cells, and not to background fluorescence). The symbol ª✽º means that the sample has been filtered in order to get
rid of unbound chimera protein. A-B, images of E. coli labelled with GFP-hadrurin fluorescent chimera protein, where labeling is lost after a filtering step. C-D,
images of E. coli labelled with GFP-pb5 fluorescent chimera protein, where labeling is also lost after a filtering step. Only the fluorescent biosensor chimera protein
GFP-colS4 (E,F) is able to maintain its strong binding to the E. coli surface after the filtration step. Also, the other two bacterial species (negative controls) are not
labelled at all (H, I). Images G and g are negative controls for E. coli, where no fluorescent biosensor chimera protein has been added, as a method to test that this
bacterial cells do not show autofluorescence under these conditions.
in its soluble form (free) pb5 wouldn't be able to have an optimal conformation to facilitate the
binding. However, in vitro studies have shown that pb5 conformation is the same as free
protein and as part of phage tail [
Based on these data, we decided to use this GFP-pb5 biosensor chimera protein as a second
candidate for specific fluorescent detection of E. coli in drinking water samples (Fig 4). To
carry this out, S. albus-pJFF1 -BS2-NP spores were cultivated in 250 mL R5A production
medium in order to get cell mass to isolate GFP-pb5 biosensor chimera protein (984 amino
acids, Fig 4A). After FPLC purification steps, a concentration of 332 μg/mL biosensor chimera
10 / 20
Fig 4. A: Hypothetical spatial structure of the fluorescent biosensor chimera protein GFP-pb5. B: Diagram showing the attachement process of the chimera
GFP-pb5 to the FhuA transporter of E. coli. PMT: photomultiplier (part of the E. coli Analyzer device in charge of detecting and amplifying the fluorescence
signal produced by the GFP domain of each biosensor chimera protein, after receiving the 395 nm excitation light from the LED source).
protein was obtained (Bradford quantification). 2 mL of GFP-pb5 biosensor chimera protein
were used for binding assays to 100 μL of each E. coli dilution. protein is a highly unstable
protein that tends to aggregate in concentrations above 500 μg/mL.
In order to visualize binding of the GFP-pb5 biosensor chimera protein to cells, the
fluorescence was measured using the E. coli Analyzer fluorimeter, but these analyses rendered very
low fluorescence signals.
Further analyses of these same samples were carried out under confocal microscope.
Confocal images showed a better GFP-pb5 labeling of cells only in samples where no filtering step
was applied. This indicated that probably, GFP-pb5 chimera protein binding to E. coli cells
was labile, and that most bound GFP-pb5 was washed away during the filtration step used to
get rid of unbound chimera protein (Fig 3C and 3D). Therefore, we decided to stop further
experiments with GFP-pb5.
GFP-colS4 biosensor chimera protein
Colicins have been extensively studied since its discovery in 1925. These plasmid-encoded
toxic proteins are synthesized by about 50% of E. coli strains and their function is to specifically
eliminate other E. coli competitive strains, under certain stress conditions [
]. The most
common toxicity mechanism for colicins is the formation of a pore in the cytoplasmic membrane,
followed by an interruption of the electrochemical gradient [
Based on this knowledge, colicin S4 was chosen as a third alternative for construction of a
new biosensor chimera protein. The sequence of this colicin shows a characteristic mosaic-like
structure consisting of three domains: a N-terminal domain responsible for translocation
11 / 20
Fig 5. A: Hypothetical spatial structure of the fluorescent biosensor chimera protein GFP-colS4. B: Diagram showing the attachement process of the fluorescent
biosensor chimera protein to the OmpW receptor of E. coli. PMT: photomultiplier (part of the E. coli Analyzer device in charge of detecting and amplifying the
fluorescence signal produced by the GFP domain of each biosensor chimera protein, after receiving the 395 nm excitation light from the LED source).
through the outer membrane by interacting with the Tol proteins system; a central domain
that mediates the binding to its OmpW receptor located in the outer membrane; and a
C-terminal domain responsible for the pore formation. This narrow host range is determined by a
highly specific binding of the colicin S4 into E. coli sensitive cells [
] (Fig 5).
The initial contact between colicin S4 and E. coli cell is established between the conserved
receptor binding domain and the OmpW specific receptor in the bacterial outer membrane
which is a member of a large family of small β-barrel proteins. After this receptor recognition,
the translocation domain can penetrate through the pores of the OmpW receptor across the
outer membrane, towards the periplasm. Once in the periplasm, the colicin acts on the plasma
membrane, generating pores [
Colicin S4 is the only colicin containing two identical receptor binding domains, both
recognizing E. coli OmpW protein, but the binding of one of them is enough for triggering an
effect on the E. coli cell [
]. OmpW is structured in eight transmembrane β-strands with
extended loops exposed to the environment [
]. The first helix from both colicin S4 receptor
binding domains (R1α1 and R2α1) are able to interact with the Asp116, His117 and Glu120
amino acids of OmpW, which are freely exposed to the extracellular space [
In order to test the new GFP-colS4 chimera protein as a new candidate for florescent
labeling of E. coli cells (Fig 5), S. albus-pLMF -BS2-NP spores were cultivated in 250 mL R5A
production medium in order to get cell mass to isolate GFP-colicin S4 biosensor chimera protein
(843 amino acids, Fig 5A). After FPLC purification steps, a concentration of 537 μg/mL
biosensor protein was obtained (measured by Bradford). 2 mL of GFP-colicin S4 protein were
used for binding assays to 100 μL of each E. coli dilution (see Materials and methods section).
12 / 20
Fig 6. Sensitivity curve using GFP-colS4 as biosensor fluorescent chimera protein for E.coli (blue dots) at diverse CFU
concentrations, from 20 to 103 CFU, which is the detected linear range. Dots represent the average values and the standard
error from three different detection experiments at each condition. Over 103 CFU the method shows saturation. Error bars
for each triplicate experiment are shown. Data for S. enterica var. arizonae (green dots) and E. cloacae (red dots) are also
In order to visualize binding of the GFP-colicin S4 biosensor chimera protein to E. coli
cells, the fluorescence was measured with the E. coli Analyzer fluorimeter, getting good
fluorescence signal values (obtained as mV output), which were used for generating the sensitivity
curve (Fig 6). This curve shows that in binding experiments with more than 3x103 E. coli CFU,
the system gets saturated, and then there is not a further increase in the generated mV output
signal. Also, with E. coli CFU below 20, the signal is also no longer proportional to the number
of E. coli CFU, as signal intensities below these 20 CFU are three times of standard deviation
lower than the mean intensity from background signal (S1 Table). Therefore, the lineal signal
range obtained with these experiments is between 20 and 3x103 E. coli CFU. (Fig 6).
We decided to analyze the same GFP-colicin S4 samples under confocal microscope, with
and without the filtering step (which is used for getting rid of the unbound chimera protein).
Confocal images showed that GFP-colicin S4 labelled E. coli cells perfectly, even after the
filtration step, a procedure which causes unbinding in the cases of GFP-hadrurin and GFP-pb5
experiments (Fig 3E and 3F).
Moreover, in order to test the specificity of this method for E. coli, the same labeling
experiments were carried out with dilutions of Enterobacter cloacae and Salmonella enterica var.
arizonae as negative controls. These experiments showed that GFP-ColS4 biosensor chimera
protein was not recognizing these other two species (Figs 6 and 3H and 3I), validating the fact
that this specificity of the GFP-colS4 chimera towards E. coli is not due to an unspecific
binding through the GFP nor the central linker peptide domains, as these ones are also present in
the two other chimeric proteins, which lack this binding specificity for E. coli.
In this work, three independent alternatives have been designed and created in order to
specifically detect the sentinel bacteria E. coli in drinking water samples. These three alternatives are
based in an easy procedure dealing with biosensor chimera proteins, where a GFP
amino13 / 20
terminal domain is responsible for emitting fluorescence (after absorbance of UV light at 395
nm in a spectrophotometer cuvette) [
]. In contrast, the carboxyl-terminal domain at each
chimera (hadrurin, pb5 or colS4 domains) is responsible for binding specifically to different
structures in the external surface of E. coli cells (membrane, or the FhuA or OmpW proteins,
respectively) [46±48]. Both domains (GFP and binding domains) are connected by a flexible
artificial peptide linker (100 amino acids long), in order to facilitate free domains movements
and to avoid steric complications between both at each chimera protein. We have carefully
designed in silico these 100 amino acids linker in order to accomplish a totally flexible arm
which is able to maintain the connectivity of both domains (GFP and binding domains) as a
way to preserve their biological activities (fluorescence and binding, respectively). This total
flexibility is shown in Figs 2, 4 and 5, and was secured by careful in silico testing for the absence
of any α-helix nor β-sheet subsections, testing its whole sequence, as well as previous fragments
and versions, for secondary structures [
The first biosensor chimera protein developed in this work, GFP-hadrurin, showed a very
low binding to E. coli cells (to its outer membrane), independently of the sample processing
(with or without filtering step used to get rid of free unbound chimera) [
]. The reason for
this low labeling could be a low integration efficiency of the chimeric protein in the bacterial
outer membrane. Another reason could be a steric inability of the biosensor chimera protein,
as the small hadrurin α-helix carboxyl terminal domain could be blocked by surrounding
linker and GFP domains. The reason for this is that the huge GFP amino terminal domain
may partially encompass this domain, avoiding its access to the outer membrane, therefore
causing low fluorescence labeling of bacterial cells.
In these and further experiments using confocal microscope, E. coli cells were always
checked to test that they did not show autofluorescence under these experimental conditions,
before using the corresponding samples with biosensor chimeras added (Fig 3G).
In a similar way, the experiments involving the second alternative, the GFP-pb5 chimera
protein, showed a labile binding of this biosensor protein to E. coli cells, as adding a filtering
step after the initial chimeric protein plus cell dilutions mix reduced significantly the obtained
fluorescence labelling. This is in contrast to previous descriptions where free pb5 protein was
able to bind to its FhuA bacterial receptor as a pure protein [55±58,65]. Purified pb5 is poorly
soluble, as in published studies it precipitates at a concentration above 0.5 mg/mL or in the
presence of imidazole [
]. Also, free pb5 protein tends to self-assembly, and these forms are
not biologically active [
]. These described characteristics on free pb5 could be responsible
for the low labelling detected in our experiments with the GFP-pb5 chimera protein, although
in our experiments, no initial evidence of aggregation exists. Therefore, another possibility is
that somehow the pb5 domain is suffering some conformational changes which makes its
binding to the bacteria not stable enough to filter the solution before proceeding to the
measure of fluorescence in the E. coli Analyzer. Further experiments with GFP-pb5 chimera were
not carried out, as the biosensor performance was better with GFP-colS4 (see below).
However, the third chimeric protein designed in this work, GFP-colS4, showed excellent
performance in our in vitro labelling tests for E. coli, by binding of its OmpW receptor in the
outer membrane, as it has been described for this colicin [
Previous studies had shown the expression of colicin S4 in a wild type strain of E. coli
bearing a plasmid which contained its gene [
]. In these experiments, colicin S4 was accumulated
in the E. coli cytosol. However, the presence of OmpW receptors in this species made these
cells sensitive to the action of this colicin S4, because a part of the cytosol expressed colicin was
released into the culture medium in low concentrations, killing the bacteria [
expression and purification of colicinS4 in E. coli by other authors forced the use of
OmpW14 / 20
mutant strains (as 5KΔmpW E. coli strain), or the simultaneous expression of the csi gene,
responsible for immunity to colicin S4 in E. coli [
However, in our expression system, these problems associated with the production of a
biosensor chimera protein based on a colicin, as colicin S4, are absent. This is due to the fact that
the bacterial factory used for expression of the fluorescent biosensor chimera protein is the
actinomycete S. albus, which is not targeted by colicins.
In our experiments with the GFP-colS4 chimera protein, the fluorescent labelling of E. coli
cells is good, even though colicin S4 domain theoretically would cause a translocation to the
periplasm, through its OmpW receptor [
]. In the case of the chimera, however, probably its
complete translocation has not taken place across the outer membrane, and part of it (at least
the GFP domain) could stay on the outer side of the bacterial cell. The reason for this
statement is that it has been described that versions of colicin S4 containing a His-tag are able to
insert itself spontaneously in the outer membrane, but less frequently than wild type colicin S4
]. Therefore, this His-tag somehow hindered the insertion into the outer membrane, but
not the binding to its receptor [
]. In our case, binding to OmpW seems to happen at high
rates, and a nice fluorescence signal is also obtained, maybe indicating that the GFP domain
has not been altered in its structure nor translocated across OmpW receptor (Fig 3). Although
colicin S4 is known to bind to E. coli and finally to cause killing of these bacteria by forming
pores in the cytoplasmic membrane, in our case, the short time for the experimental procedure
and the fact that we actually see the E. coli cells under confocal microscope after labeling them
with GFP-colS4 magic bullet, supports the idea that the eventual killing of these bacteria is not
affecting our sensor approach. Moreover, our fast experimental design is not affected by the
eventual killing of the bacterial cells after GFP-colS4 binding, since once the magic bullet is
bound, we will get back the corresponding fluorescence from labelled cells.
OmpW expression in E. coli cells varies significantly in response to various environmental
signals, such as temperature [
]. So, E. coli cells have shown greater resistance to colicin S4
when they grow at 23ÊC, compared to 37ÊC. The reason for this is a sharp reduction or even
absence of OmpW receptor content at low temperatures as 23ÊC, as temperature regulates
ompW gene expression at the transcriptional level [
]. This feature gives greater specificity to
the method described in this study, as it would ensure that the E. coli contamination detected
in the drinking water samples would be due to a recent faecal contamination (with high
expression of OmpW protein as these cells were growing at 37ÊC), and not to a possible
multiplication of the microorganism in the environment at low temperatures. Further studies in
this sense will be carried out.
Three different biosensor chimera proteins have been designed and tested for specific binding
to E. coli cells in drinking water samples, in order to achieve an easy and fast detection method
based on fluorescence. Each one of the designed biosensor chimera proteins contains a GFP
domain placed at the amino terminus, in charge of emitting fluorescence once irradiated with
UV light at 395 nm. Also, a magic bullet domain, for specific binding to some E. coli external
structures (membrane, or FhuA and OmpW proteins) is present at the carboxyl terminus of
each biosensor chimeric protein. Both important domains are linked through a 100 amino
acids unfolded section, in order to ensure that no steric problems arise between them, and that
the domain in charge of binding to the bacterial cell is freely moving with respect to the GFP
domain. Selection of each one of the three binding domains (at the carboxyl-terminal of each
chimera protein) has been carried out based on abundant literature on the binding specificities
and requirements for each binding subunit: hadrurin is a simple and small lineal α-helix
15 / 20
Fig 7. MUSCLE multiple sequence alignment of the OmpW proteins from E. coli, S. enterica var. arizonae and E. cloacae, highlighting in yellow the region
encompasing the the Asp116, His117 and Glu120 amino acids of OmpW (in blue letters: D, H and E), which are freely exposed to the extracellular space and
act as binding motifs for colicin S4 bacteriocin (as well as for GFP-colS4 chimera protein in this biosensor). As it is shown, the other two enterobacteria species
do not have the conserved motif DHÐE as in E. coli. Accesion numbers: SAI89619 (E. cloacae), BAA14788 (E. coli), OSE54138 (S. enterica).
protein binding to the outer membranes of Enterobacteriaceae members; pb5 is part of the
recognition structure of the central tail fibers near the base plate of T5 phage, in charge of initial
binding to outer membrane iron ferrichrome transporter FhuA in E. coli, and colicin S4 is a
bacteriocin with high specificity for E. coli outer membrane protein OmpW.
Although some unspecific binding properties could arise due to the global structure of each
one of the three chimera proteins, due to the presence of the amino-terminal GFP domain or
to the central flexible synthetic peptide linker (showing a structure of a random coil lacking
αhelixes nor β-sheets), other tested bacterial species as S. enterica and E. cloacae clearly ruled out
this unspecificity, validating the initial experimental approach. Remarkably, these two other
enterobacteria species do not have the conserved motif DHxxE as in E. coli (Fig 7). These
three amino acids are the binding ones for colicin S4, as it has been described above [
The whole method relies also in the use of a specific a la carte portable detection device
(E. coli Analyzer) which contains a LED source, a detection chamber, a photomultiplier and a
converter of the fluorescence signal into mV data.
Two of these biosensor chimera proteins, the GFP-hadrurin and the GFP-pb5, showed a
non-optimal performance with respect to binding parameters to E. coli, as the fluorescence
signals obtained after the filtration step (necessary to get rid of unbound biosensor protein) were
very low, probably due to some structural problems at the binding domains. However, the
GFP-colS4 biosensor chimera protein showed nice stability and binding parameters, and was
able to carry out the specific detection of E. coli with a linear range from 20 to 103 CFU, and a
lower detection level of 20 CFU in just 8 min, which are quite interesting features for its future
development as a commercial detection method.
S1 Table. Detection signals obtained from 20 to 1000 CFUs for the three pathogens used in
this study, using the E. coli Bioanalyzer device developed in the study.
16 / 20
We thank financial support from FICYT (FundacioÂn para el Fomento en Asturias de la
InvestigacioÂn CientÂõfica Aplicada y la TecnologÂõa) and IDEPA (Instituto de Desarrollo EconoÂmico
del Principado de Asturias) through the Projects IE09-106 and IDE2012/000452 to HIPSITEC
SA. We also thank Servicios CientÂõfico-TeÂcnicos from the University of Oviedo for technical
support in protein purification.
Conceptualization: MarÂõa AÂ lvarez San MillaÂn, Natalia CobiaÂn, Felipe LomboÂ.
Funding acquisition: Natalia CobiaÂn, Felipe LomboÂ.
Investigation: Ignacio GutieÂrrez-del-RÂõo, Laura MarÂõn, Javier FernaÂndez, MarÂõa AÂ lvarez San
MillaÂn, Natalia CobiaÂn.
Methodology: Francisco Javier Ferrero, Marta Valledor, Juan Carlos Campo, Natalia CobiaÂn,
Software: Francisco Javier Ferrero, Marta Valledor, Juan Carlos Campo, Ignacio MeÂndez.
Supervision: Francisco Javier Ferrero, Natalia CobiaÂn, Felipe LomboÂ.
Validation: Ignacio GutieÂrrez-del-RÂõo, Laura MarÂõn, Javier FernaÂndez.
Writing ± original draft: Ignacio GutieÂrrez-del-RÂõo, Laura MarÂõn, Javier FernaÂndez, Francisco
Javier Ferrero, Felipe LomboÂ.
Writing ± review & editing: Felipe LomboÂ.
17 / 20
18 / 20
Kim M, Jung T, Kim Y, Lee C, Woo K, Hun J, et al. Biosensors and Bioelectronics Short communication
A micro fl uidic device for label-free detection of Escherichia coli in drinking water using positive
dielectrophoretic focusing, capturing, and impedance measurement. 2015; 74: 1011±1015.
WHO and UNICEF. 2015 Update and MDG Assessment. 2015;
COUNCIL DIRECTIVE 98/83/EC of 3 November 1998 on the quality of water intended for human
21 CFR 129ÐPROCESSING AND BOTTLING OF BOTTLED DRINKING WATER. https://www.gpo.
Edberg SC, Rice EW, Karlin RJ, Allen MJ. Escherichia coli: the best biological drinking water indicator
for public health protection. Symp Ser Soc Appl Microbiol. 2000; 106S±116S. PMID: 10880185
Edberg SC, Rice EW, Karlin RJ, Allen MJ. Escherichia coli: the best biological drinking water indicator
for public health protection. Symp Ser Soc Appl Microbiol. 2000; 106S±116S. Available: http://www.
19 / 20
WHO WHO. OMS | Water [Internet]. World Health Organization; 2014 .
2. Gibson KE . Viral pathogens in water: occurrence, public health impact, and available control strategies . Curr Opin Virol . 2014 ; 4 : 50 ±7. https://doi.org/10.1016/j.coviro. 2013 . 12 .005 PMID: 24440908
3. Nygård K , Schimmer B , Søbstad Ø , Walde A , Tveit I , Langeland N , et al. A large community outbreak of waterborne giardiasis-delayed detection in a non-endemic urban area . BMC Public Health . 2006 ; 6 : 141 . https://doi.org/10.1186/ 1471 -2458-6-141 PMID: 16725025
4. Nichols GL . Food-borne protozoa . Br Med Bull . 2000 ; 56 : 209 ± 35 . http://www.ncbi.nlm.nih.gov/ pubmed/10885117
5. Lee MB . Everyday and exotic foodborne parasites . Can J Infect Dis . 2000 ; 11 : 155 ±8. http://www.ncbi. nlm.nih.gov/pubmed/18159283
6. Lupidi R , Cinco M , Balanzin D , Delprete E , Varaldo PE . Serological follow-up of patients involved in a localized outbreak of leptospirosis . J Clin Microbiol . 1991 ; 29 : 805 ± 9 . Available: http://www.ncbi.nlm. nih.gov/pubmed/1890181
7. Liang JL , Dziuban EJ , Craun GF , Hill V , Moore MR , Gelting RJ , et al. Surveillance for waterborne disease and outbreaks associated with drinking water and water not intended for drinkingÐUnited States , 2003 ± 2004 . MMWR Surveill Summ. 2006 ; 55 : 31 ± 65 . Available: http://www.ncbi.nlm.nih.gov/pubmed/ 17183231
8. Jeandron A , Saidi JM , Kapama A , Burhole M , Birembano F , Vandevelde T , et al. Water supply interruptions and suspected cholera incidence: a time-series regression in the Democratic Republic of the Congo . PLoS Med . 2015 ; 12 : e1001893. https://doi.org/10.1371/journal.pmed. 1001893 PMID: 26506001
9. Tallon LA , Love DC , Moore ZS , Sobsey MD . Recovery and sequence analysis of hepatitis a virus from springwater implicated in an outbreak of acute viral hepatitis . Appl Environ Microbiol . 2008 ; 74 : 6158 ± 60 . https://doi.org/10.1128/AEM.02872-07 PMID: 18708522
10. Cabral JPS . Water microbiology. Bacterial pathogens and water . Int J Environ Res Public Health . 2010 ; 7 : 3657 ± 703 . https://doi.org/10.3390/ijerph7103657 PMID: 21139855
Toxicology . 2004 ; 198 : 229 ± 38 . https://doi.org/10.1016/j.tox. 2004 . 01 .030 PMID: 15138046
Stauber C , Miller C , Cantrell B , Kroell K . Evaluation of the compartment bag test for the detection of Escherichia coli in water . J Microbiol Methods . 2014 ; 99 : 66 ± 70 . https://doi.org/10.1016/j.mimet. 2014 .
02.008 PMID: 24566129
WHO WHO . Combating waterborne disease at the household [Internet]. 2007 pp. 4 ± 36 .
Golberg A , Linshiz G , Kravets I , Stawski N , Hillson NJ , Yarmush ML , et al. Cloud-enabled microscopy and droplet microfluidic platform for specific detection of Escherichia coli in water . PLoS One . 2014 ; 9 : 4± 12 . https://doi.org/10.1371/journal.pone. 0086341 PMID: 24475107
WHO WHO . Guidelines for Drinking-water Quality . Third. Switzerland: OMS Editions; 2008 .
Anderson KL , Whitlock JE , Harwood VJ . Persistence and Differential Survival of Fecal Indicator Bacteria in Subtropical Waters and Sediments . Appl Environ Microbiol . 2005 ; 71 : 3041 ± 3048 . https://doi.org/ 10.1128/AEM.71.6. 3041 - 3048 . 2005 PMID: 15933000
WHO WHO . Assessing Microbial Safety of Drinking Water . IWA Publis . London; 2003 .
Chen J , Alcaine SD , Jiang Z , Rotello VM , Nugen SR . Detection of Escherichia coli in Drinking Water Using T7 Bacteriophage-Conjugated Magnetic Probe . Anal Chem . 2015 ; 87 : 8977 ± 8984 . https://doi.
org/10 .1021/acs.analchem. 5b02175 PMID: 26172120
26. Juhna T , Birzniece D , Larsson S , Zulenkovs D , Sharipo A , Azevedo NF , et al. Detection of Escherichia coli in biofilms from pipe samples and coupons in drinking water distribution networks . Appl Environ Microbiol . 2007 ; 73 : 7456 ± 64 . https://doi.org/10.1128/AEM.00845-07 PMID: 17720845
Liu Y , Gilchrist A , Zhang J , Li X-F . Detection of viable but nonculturable Escherichia coli O157:H7 bacteria in drinking water and river water . Appl Environ Microbiol . 2008 ; 74 : 1502 ±7. https://doi.org/10.1128/ AEM.02125-07 PMID: 18203853
2009 ; 24 : 3365 ± 71 . https://doi.org/10.1016/j.bios. 2009 . 04 .047 PMID: 19481924
Rompr eÂ A , Servais P , Baudart J , De-Roubin MR , Laurent P . Detection and enumeration of coliforms in drinking water: current methods and emerging approaches . J Microbiol Methods . 2002 ; 49 : 31 ± 54 .
Lim D V , Simpson JM , Kearns EA , Kramer MF . Current and developing technologies for monitoring agents of bioterrorism and biowarfare . Clin Microbiol Rev . 2005 ; 18 : 583 ± 607 . https://doi.org/10.1128/ CMR.18.4. 583 - 607 . 2005 PMID: 16223949
Odumeru JA , Steele M , Fruhner L , Larkin C , Jiang J , Mann E , et al. Evaluation of accuracy and repeatability of identification of food-borne pathogens by automated bacterial identification systems . J Clin Microbiol . 1999 ; 37 : 944 ± 9 . Available: http://www.ncbi.nlm.nih.gov/pubmed/10074506
Zhao X , Hilliard LR , Mechery SJ , Wang Y , Bagwe RP , Jin S , et al. A rapid bioassay for single bacterial cell quantitation using bioconjugated nanoparticles . Proc Natl Acad Sci U S A . 2004 ; 101 : 15027 ± 32 .
https://doi.org/10.1073/pnas.0404806101 PMID: 15477593
2006 ; 72 : 228 ± 32 . https://doi.org/10.1128/AEM.72.1. 228 - 232 . 2006 PMID: 16391047
34. Koo OK , Liu Y , Shuaib S , Bhattacharya S , Ladisch MR , Bashir R , et al. Targeted capture of pathogenic bacteria using a mammalian cell receptor coupled with dielectrophoresis on a biochip . Anal Chem . 2009 ; 81 : 3094 ± 101 . https://doi.org/10.1021/ac9000833 PMID: 19317455
35. Chen J , Jiang Z , Ackerman JD , Yazdani M , Hou S , Nugen SR , et al. Electrochemical NanoparticleEnzyme Sensors for Screening Bacterial Contamination in Drinking Water . Analyst. 2013 ; 5 ± 10 . https:// doi.org/10.1039/C5AN00637F PMID: 26042607
36. Ivnitski D , Abdel-Hamid I , Atanasov P , Wilkins E . Biosensors for detection of pathogenic bacteria . Biosens Bioelectron . 1999 ; 14 : 599 ± 624 . https://doi.org/10.1016/S0956- 5663 ( 99 ) 00039 - 1
37. Lim D V , Simpson JM , Kearns EA , Kramer MF . Current and Developing Technologies for Monitoring Agents of Bioterrorism and Biowarfare . 2005 ; 18 : 583 ± 607 .
38. Aguirrezabalaga I , Olano C , Allende N , Rodriguez L , Braña AF , MeÂndez C , et al. Identification and expression of genes involved in biosynthesis of L-oleandrose and its intermediate L-olivose in the oleandomycin producer Streptomyces antibioticus . Antimicrob Agents Chemother . 2000 ; 44 : 1266 ± 75 . PMID: 10770761
39. Kieser T , Chater K , Bibb M , Buttner M , Hopwood D. Practical Streptomyces Genetics. England : J. 2000 .
40. Sambrook J , Russell D W. Molecular Cloning : A Laboratory Manual . Cold Spring Harbor Laboratory Press, Cold Spring Harbor, NY. 2001 .
41. FernaÂndez E , Weissbach U , Reillo S , Braña F , MeÂndez C , Rohr J , et al. Identification of two genes from Streptomyces argillaceus encoding glycosyltransferases involved in transfer of a disaccharide during biosynthesis of the antitumor drug mithramycin . J Bacteriol . 1998 ; 180 : 4929 ± 4937 . PMID: 9733697
42. Ferrero Martin FJ , Valledor Llopis M , Campo Rodriguez JC , Marin Fernandez L , Gutierrez-del-Rio Menendez I , Fernandez Fernandez J , et al. A Novel Handheld Fluorimeter for Rapid Detection of Escherichia coli in Drinking Water . IEEE Sens J. 2016 ; 16 : 5136 ± 5144 . https://doi.org/10.1109/JSEN. 2016 .2558649
43. Musienko Y , Reucroft S , Swain J. The gain, photon detection efficiency and excess noise factor of multi-pixel Geiger-mode avalanche photodiodes . Nucl Instruments Methods Phys Res Sect A Accel Spectrometers, Detect Assoc Equip . 2006 ; 567 : 57 ± 61 . https://doi.org/10.1016/j.nima. 2006 . 05 .214
44. Tsien RY . The green fluorescent protein . Annu Rev Biochem . 1998 ; 67 : 509 ± 44 . https://doi.org/10. 1146/annurev. biochem.67.1.509 PMID: 9759496
45. International Organization for Standardization. ISO 9308 ± 1:2014ÐWater qualityÐEnumeration of Escherichia coli and coliform bacteria . 2014 .
46. Torres-Larios A , Gurrola GB , Zamudio FZ , Possani LD . Hadrurin, a new antimicrobial peptide from the venom of the scorpion Hadrurus aztecus . Eur J Biochem . 2000 ; 267 : 5023 ± 31 . PMID: 10931184
47. Mondigler M , Holz T , Heller KJ . Identification of the receptor-binding regions of pb5 proteins of bacteriophages T5 and BF23 . Virology. 1996 ; 219 : 19 ± 28 . https://doi.org/10.1006/viro. 1996 .0218 PMID: 8623528
48. Arnold T , Zeth K , Linke D . Structure and function of colicin S4, a colicin with a duplicated receptor-binding domain . J Biol Chem . 2009 ; 284Arnold,: 6403 ± 13 . https://doi.org/10.1074/jbc.M808504200 PMID: 19056731
49. Pilsl H , Smajs D , Braun V . Characterization of Colicin S4 and Its Receptor, OmpW, a Minor Protein of the Escherichia coli Outer Membrane Characterization of Colicin S4 and Its Receptor, OmpW, a Minor Protein of the Escherichia coli Outer Membrane . J Bacteriol . 1999 ; 181 : 3578 ± 80 . PMID: 10348872
50. SaÂnchez-VaÂsquez L , Silva-Sanchez J , JimeÂnez-Vargas JM , RodrÂõguez-Romero A , Muñoz-Garay C , RodrÂõguez MC , et al. Enhanced antimicrobial activity of novel synthetic peptides derived from vejovine and hadrurin . Biochim Biophys Acta . 2013 ; 1830 : 3427 ± 36 . https://doi.org/10.1016/j.bbagen. 2013 . 01 . 028 PMID: 23403131
51. PlancËon L , Janmot C , le Maire M , Desmadril M , Bonhivers M , Letellier L , et al. Characterization of a High-affinity Complex Between the Bacterial Outer Membrane Protein FhuA and the Phage T5 Protein pb5 . J Mol Biol . 2002 ; 318 : 557 ± 569 . https://doi.org/10.1016/S0022- 2836 ( 02 ) 00089 -X PMID: 12051859
52. Braun V. FhuA (TonA), the career of a protein . J Bacteriol . 2009 ; 191 : 3431 ±6. https://doi.org/10.1128/ JB.00106-09 PMID: 19329642
53. Sayers JR . Bacteriophage T5 . In: Calendar R, editor. The Bacteriophages . 2005 .
54. Endriss F , Braun V . Loop deletions indicate regions important for FhuA transport and receptor functions in Escherichia coli . J Bacteriol . 2004 ; 186 : 4818 ± 23 . https://doi.org/10.1128/JB.186.14. 4818 - 4823 . 2004 PMID: 15231815
55. Killmann H , Videnov G , Jung G , Schwarz H , Braun V . Identification of receptor binding sites by competitive peptide mapping: phages T1, T5, and phi 80 and colicin M bind to the gating loop of FhuA . J Bacteriol . 1995 ; 177 : 694 ± 698 . PMID: 7836303
56. Flayhan A , Wien F , Paternostre M , Boulanger P , Breyton C . New insights into pb5, the receptor binding protein of bacteriophage T5, and its interaction with its Escherichia coli receptor FhuA . Biochimie . 2012 ; 94 : 1982 ± 1989 .
57. Breyton C , Flayhan A , Gabel F , Lethier M , Durand G , Boulanger P , et al. Assessing the Conformational Changes of pb5, the Receptor-binding Protein of Phage T5, upon Binding to Its Escherichia coli Receptor FhuA . J Biol Chem . 2013 ; 288 : 30763 ± 30772 . https://doi.org/10.1074/jbc. M113.501536 PMID: 24014030
58. Mondigler M , Viigele RT , Heller KJ . Overproduced and purified receptor binding protein pb5 of bacteriophage T5 binds to the T5 receptor protein FhuA . 1995 ; 130 : 293 ± 300 .
59. Lakey JH , Slatin SL . Pore-forming colicins and their relatives . Curr Top Microbiol Immunol . 2001 ; 257 : 131 ± 61 . PMID: 11417119
60. Cascales E , Buchanan SK , DucheÂ D , Kleanthous C , Lloubès R , Postle K , et al. Colicin biology . Microbiol Mol Biol Rev . 2007 ; 71 : 158 ± 229 . https://doi.org/10.1128/MMBR.00036-06 PMID: 17347522
61. Housden NG , Loftus SR , Moore GR , James R , Kleanthous C . Cell entry mechanism of enzymatic bacterial colicins: porin recruitment and the thermodynamics of receptor binding . Proc Natl Acad Sci U S A . 2005 ; 102 : 13849 ± 54 . https://doi.org/10.1073/pnas.0503567102 PMID: 16166265
62. McClean S. Eight stranded β -barrel and related outer membrane proteins: role in bacterial pathogenesis . Protein Pept Lett . 2012 ; 19 : 1013 ± 25 . PMID: 22533621
63. Hong H , Patel DR , Tamm LK , van den Berg B. The outer membrane protein OmpW forms an eightstranded beta-barrel with a hydrophobic channel . J Biol Chem . 2006 ; 281 : 7568 ± 77 . https://doi.org/10. 1074/jbc.M512365200 PMID: 16414958
64. Chen X , Zaro JL , Shen W -C. Fusion protein linkers: property, design and functionality . Adv Drug Deliv Rev . 2013 ; 65 : 1357 ± 69 .
65. Brambilla L , MoraÂn-Barrio J , Viale AM . Expression of the Escherichia coli ompW colicin S4 receptor gene is regulated by temperature and modulated by the H-NS and StpA nucleoid-associated proteins . FEMS Microbiol Lett . 2014 ; 352 : 238 ± 244 . https://doi.org/10.1111/ 1574 - 6968 .12385 PMID: 24444297